The Dynamic Arc is the premier functional balance system utilized in healthcare today. Whether for balance screening, assessment or rehabilitation/treatment, the Dynamic Arc provides an objective platform to provide optimal care to your patients and residents.
This brief video will give you insight to the specific role that the Dynamic Arc will have in the provision of your services.
If your product came with the included CD-Rom, insert the disc into your laptop at this time. Click on the disc drive to open the software then complete the following steps:
TO LOAD THE SOFTWARE ONTO YOUR COMPUTER:
WORKING WITH THE BARNETT BALANCE ASSESSMENT PROGRAM:
The program itself does not store patient information but rather acts as a collection and scoring platform. In the program there is 2 tabs at the top; the "Worksheet" and the "Results". You will input all the appropriate data into the worksheet, and then click the results tab upon completion for normative comparisons and raw scores. Once all data is entered and results tallied, you can either print the score sheets to your local or network-based printer or save the image as a .tiff to your desktop to load into your EMR. The printed score sheets (there will be 2 in total) can also be scanned and uploaded to your EMR should you choose that option as well.
Make sure you maintain all files in the designated folders or the program will not work correctly.
The Dynamic Arc has 2 components for assembly, the base/column and the arc itself. To assemble, raise the column up and fasten screw to secure. Make sure that the screw is tight to prevent sway in the column itself.
Then, place the arc on to the column with the post placed in the hole on top. Make sure the arc is properly aligned with the etching on the moving mechanism placed in the front side of the column. Also, make sure that the screw on the underside of the Dynamic Arc rests in the groove that is on the collar of the column.